﻿ Understanding accuracy specifications - May 2019 - Digitial Industries - Process Automation Solutions - SA Instrumentation & Control

#### Understanding accuracy specifications

May 2019 Pressure Measurement & Control

Pressure measurement device accuracies are commonly specified as percentage of full scale, or percentage of reading, and the difference is significant. If an accuracy statement simply names a percentage (0,1%), it is normally specifying a percentage of full scale.

Accuracy (or really inaccuracy) of pressure instruments is therefore specified in one of two ways:

• Accuracy as a percentage of full scale(% of FS).

• Accuracy as a percentage of reading (% of RD).

If an instrument has an accuracy specified as % of FS, the error will have a fixed value, no matter where the applied pressure is in the measuring sensor’s range. Take, for example, an instrument calibrated for a pressure range of 0-100 kPa, with a stated accuracy of 0,1% of FS. At an applied pressure of 100 kPa (full scale of the instrument) the error will be at most 0,1% of full scale, or ±0,1 kPa.

As the pressure moves down, the error will remain 0,1% of FS. Therefore, at half scale (50 kPa), that error becomes proportionally larger (0,2%). The further down from full scale, the further the accuracy decreases. At 10 kPa the error is 1% and at 1 kPa the error is 10%, which still relates to a ±0,1 kPa inaccuracy.

If an instrument has accuracy specified as 0,1% of RD, the error will always be the same percentage of the actual applied pressure. Using the same example as above of 0-100 kPa, but this time, with a stated accuracy of 0,1% RD, at 10 kPa the error is ±0.01 kPa i.e. still 0,1% accuracy, and 10 times more accurate than an instrument specified as % of FS.

Table 1.

In summary, as per Table 1, 0,1% of FS accuracy is ±0,1 kPa across the scale. With a turndown of 10:1, the % error becomes 10 times as large at the lower end.

In comparison, for 0,1% of RD, the accuracy remains 0,1%, no matter the turndown ratio. However, it is important to note that this cannot continue all the way to zero, as there is a physical limit to how well the sensor can perform.

The latest Schneider Electric IGP50S gauge pressure transmitter range – also available as absolute or differential variants – with FoxCal technology (11 pre-calibrated ranges imbedded in the sensor), are leaders with their stated accuracy of 0,025% of RD and very high stability of <±0,015% of URL (upper range limit) per year.

The 400:1 turn-down ratio, coupled with the built-in calibration curves, allows for a single sensor to cover most ranges of pressure measurement with a single transmitter, resulting in drastically reduced inventory, engineering and configuration time. The robust design includes a standard five-year warranty and SIL2 safety certification.

For more information contact Johan van Jaarsveldt, EOH, +27 87 803 9783, [email protected], www.eoh-pas.co.za

Credit(s)

 Tel: +27 11 607 8100 Email: [email protected] www: www.di-pas.co.za Articles: More information and articles about Digitial Industries - Process Automation Solutions

VEGA handles the pressures of water treatment systems
VEGA Controls SA Pressure Measurement & Control
A water treatment system for a major metropolitan area in the Midwestern United States demands careful monitoring and management of processes across its sprawling network. Choosing VEGA for its process automation needs meant more than just obtaining precise and reliable pressure sensors.

Greener mining through water management strategies
VEGA Controls SA Pressure Measurement & Control
Water scarcity is a pressing concern in South Africa, leading to the establishment of stricter regulations on water discharge and management. To promote greener mining operations that minimise environmental impact and optimise resource utilisation, responsible dewatering practices are essential.

Pressure sensors for hydrogen applications
Instrotech Pressure Measurement & Control
Keller manufactures isolated piezoresistive pressure sensors. The great advantage of the piezoresistive measuring principle is its high stability.

A new approach to milk processing
VEGA Controls SA Pressure Measurement & Control
Ensuring the quality and safety of milk throughout the production process is of the utmost importance in the dairy industry. Process instrumentation plays a vital role in this stage by providing accurate measurements of level, pressure and point level to guarantee optimal storage conditions.

OEM high-pressure transducer with maximum long-term stability
Instrotech Pressure Measurement & Control
Instrotech is offering KELLER 10LHP, the flagship OEM pressure transducer in the KELLER product portfolio, which exemplifies the highest standards.

Enhancing beer brewing precision
VEGA Controls SA Pressure Measurement & Control
Brewing beer is a complex process that requires utmost precision in every step. To ensure precision, it is crucial to measure and control the pressure and point levels in the wort kettle and hops dissolver. Using VEGA instrumentation can offer numerous benefits to breweries.

Sensor technology for cartonboard machine for paper and packaging
VEGA Controls SA Pressure Measurement & Control
Paper and packaging specialist, LEIPA undertook a comprehensive modernisation project, combining three stock preparation lines into one, and making investments in new plant components and extensive automation technology. Because of the good experiences the company had already had with VEGA sensors, those responsible for planning and maintenance decided in favour of the VEGABAR 82 pressure transmitter as the standard instrument for level and pressure measurement in the new stock preparation system.

Pressure measurement without risk of contamination
WIKA Instruments Pressure Measurement & Control
Injectables are among the most sensitive products in the pharmaceutical industry. Sterility is essential during production. For this reason, a manufacturer of biopharmaceuticals decided to implement the pressure monitoring of its processes with hygienic instruments from WIKA.

Pressure monitoring for oil and water tempering machines
Instrotech Pressure Measurement & Control
Temperature plays a significant role in many manufacturing processes. In such cases, a tempering machine is used to maintain the manufacturing process at the desired temperature. Keller supplied PR-21Y piezoresistive pressure transmitters to a client specialising in developing and producing oil and water tempering machines.

How to look after your pressure gauge
SA Gauge Editor's Choice Pressure Measurement & Control
Pressure and temperature gauges are very widely used on process plants. Specialising in the design and manufacture of pressure and temperature measuring instruments, SA Gauge believes that all process and ambient factors should be considered when selecting and installing pressure measuring devices. The company recommends that the following guidelines on selection and calibration be followed in order to prevent gauge failure.